# Van-der-Pauw measurement

The Van der Pauw measurement method, named after its inventor Leo J. van der Pauw, is a widely used as a 4-point measurement for determining the sheet resistance and the Hall coefficient of materials.

In 1958 van der Pauw solved the general problem of potential distribution in a thin, conductive layer of any shape and thus made it possible to carry out Hall and resistance measurements on them. However, some prerequisites must be met for a correct measurement. The thickness *t *of the sample must be homogeneous and small in relation to the distance between the contacts. It must consist of a continuous form in the mathematical sense and must therefore not have any holes or islands made of highly conductive material. The four contacts must be on the edge of the sample and small in relation to the area of the sample.

If all the requirements are met, a sample prepared with the edge contacts A, B, C and D is obtained, as shown in Fig. 2a-b. To calculate the electrical conductivity, horizontal and vertical Van der Pauw resistances *R _{(AB,CD)}*, must be measured, whereby the following applies to the measurement of the horizontal resistance:

Where *V _{C}* is the voltage of the contact measured against ground and

*I*is the current applied between the contacts A and B, starting at A and ending at B. If the contacts are changed cyclically, the vertical Van der Pauw resistance can be determined. If the horizontal and vertical resistance have been measured, the surface resistance or, if the layer thickness is known, the resistivity can be calculated by numerically solving the Van der Pauw formula:

_{AB}As the electrical conductivity is the reciprocal of the specific resistance , it can be calculated according to the following equation.

To determine the Hall coefficient of the sample, an external magnetic field with the flux density *B _{Z}* can be applied perpendicular to the surface of the sample and the change in the diagonal Van der Pauw resistance (see Fig. 3), depending on the magnetic field strength, is measured.

In the case, the following is valid:

such as

with the Hall coefficient A_{H} and the magnetic flux density B_{Z}. Offset effects in current and voltage measurement can be suppressed according to the principle of measurement direction inversion. Offset effects coming from the magnetic field strength measurement, are suppressed by measuring the diagonal voltage V_{Hall }at different magnetic field strengths and determining the Hall coefficient from the slope of the Hall voltage over magnetic field strength. The final calculation is done according to the following equation:

## What properties are determined?

The Van-der-Pauw method is used to determine the electrical transport properties of materials (bulk and thin films) like the electrical conductivity, the resistivity, the hall constant, the charge carrier concentration and the charge carrier mobility.